Command Group: Motion
NT - Set the stabilisation time for the [in motion] band
read 18,11 write 31,11
The motion detection facility provides a means of disabling certain functions whenever a condition of instability, or “motion”, is detected. The “no-motion”, or “stable” condition is achieved whenever the signal is steady for the period of time set by NT, during which it cannot fluctuate by more than NR increments.The stable condition activates the relevant bit of responses to “Info Status” (IS).
The following functions are disabled if motion is detected: “Calibrate Zero” (CZ) “Calibrate Gain” (CG) “Set Zero” (SZ) and “Set Tare” (ST).
NT This sets the time (in milliseconds) over which the weight signal is checked to see if it is “stable” or has “no-motion”. The weight signal has to vary by less than NR divisions over the time period NT to be considered “stable”.
See also NR, (IS, CZ, CG, SZ, ST).