Command Group: Motion
NR - Set the [no-motion] range
read 18,10 write 31,10
The motion detection facility provides a means of disabling certain functions whenever a condition of instability, or “motion”, is detected. The “no-motion”, or “stable” condition is achieved whenever the signal is steady for the period of time set by NT, during which it cannot fluctuate by more than NR increments.The stable condition activates the relevant bit of responses to “Info Status” (IS).
The following functions are disabled if motion is detected: “Calibrate Zero” (CZ) “Calibrate Gain” (CG) “Set Zero” (SZ) and “Set Tare” (ST).
NR This is the range within which the weighing signal is allowed to fluctuate and still be considered as “stable”.
See also NT, (IS, CZ, CG, SZ, ST).