Command Group: Calibration
AG - Set the absolute gain point
read 3,1 write 4,1
AG The command AG is used as absolute gain (or measuring range) for all weight calculations and will setup in mV/V. Permitted values are ± 32000 (= ± 3.2000 mV/V).
Factory default: 20000 @ 2.0000 mV/V input signal.
The serial interface offers two parameters: “mV/V” and “value”.
When calibration is done over the network the second value is taken from the CV register,
except System80/MCS-64 Gateways (for LDM88) that takes the second parameter from the CG value (normally 20000).
See also CE, CM, CI, DS, DP, CZ, CG, ZT, AZ, CS, FD, UR,