Command Group: Calibration
CE - Set the calibration functions to the enabled state
read 3,3 write 4,3
CE Set the calibration functions to the enabled state. This command must be issued PRIOR to any attempt to set the calibration parameters CM, CI, MR, DS, DP, CZ, CG, ZT, ZR, FD, LC, LN or CS. Issuing the command without any parameters results in the response E+XXXXX where XXXXX is the Traceable Access Code (TAC). This is an internal code that is used to record any changes in the calibration settings of the device. This is a critical feature that is required for “approved” applications, as it provides for the control of access to any command that has the potential of changing the weigher calibration value.
See also CE, CI, DS, DP, CZ, CG, ZT, AZ, CS, FD, UR,